Portable Coating Thickness Gauge KCT100 is a compact, integrated coating gauge measures non – magnetic coatings (paints, plastics, rubber, enamel …) on ferrous metals( steel / iron)
The portable coating thickness gauge uses Ruby probe: The abrasion resistance of the contact point of the probe directly affects the measurement accuracy. Ruby probe is good.
KCT100 widely used in manufacturing industry, metal processing, chemical industry, inspection institution.
• Integrated design, Ideal instrument for mobile and quick measurement.
• Ruby Probe ,wear-resisting
• One key and one hand operating ,( zeroing, calibration and measurement in one key)
• Two-Point Calibration
How to make Two-Point Calibration in the KCT100 device
Keystroke three times, enter into the process of tow-point user calibration;
when USER CAL BASE is shown on the LCD, please conduct a measurement on the base first; when S1 is shown on the LCD, place Shim S1 on the base, conduct a measurement on shim S1 and lift gauge;
when S2 is shown on the LCD, place Shim S2 on the base, conduct a measurement on shim S2 and lift gauge;
When DONE is displayed, it is ready for use.
Note: The two-point user calibration process could be repeated for better results.
|Accuracy||±[(1%~3%)H+1μm] Note: H is thickness reading|
|Smallest Measuring Area||Φ15mm|
|Smallest Curvature||for convex surface: 5mm for concave surface: 25mm|
|Smallest Base material thickness||F: 0.2mm|
|Power||AAA battery 4 pcs (Non-aviation transport)|
About Magnetic induction measurement principle
The Portable Coating Thickness Gauge KCT100 adopts Magnetic induction measurement principle.
the thickness of the coating is measured by the size of the magnetic flux flowing from the probe through the non-ferromagnetic coating into the ferromagnetic substrate. The size of the corresponding magnetoresistance can also be measured to indicate the thickness of the coating. The thicker the coating is, the magnetoresistance the larger and the smaller the magnetic flux. in principle,the thickness gauge can measures the thickness of the non-magnetic coating on the magnetic substrate.
The magnetic principle thickness gauge can be used to accurately measure the paint layer on the surface of steel, porcelain, enamel protective layer, plastic, rubber coating, various non-ferrous metal plating layers including chromium, and various anti-corrosion coatings for chemical and petroleum industries. Floor.
|KCT100 Host||1 PC|
|Standard Calibration Piece||1 SET|
|Steel substrate||1 PC|
|User Manual||1 PC|
|Instrument Case||1 PC|
Quality First, Safety Guaranteed